NSF/ANSI 49 and BMBL (Biosafety in Microbiological and Biomedical Laboratories) reference the term “Relocated” as the measure of requiring re-certification. Vertical up and down BSC movement on a motorized base stand is not relocation as intended by either of these documents. The height adjustment up and down movement does not cause the vibration necessary to harm or breach the integrity of the HEPA filters. The height adjustment up and down movement is a very slow and smooth movement typical driven by a hydraulic system. If the height adjustment is used in conjunction with an exhausted BSC (i.e. type B1/B2 or canopy connected type A2), the site assessment certification test should include airflow verification throughout the range of motion. The duct connection itself should be made with flexible PVC duct material.
[Bulletin] Concurrent Balance Value of Class II, Type B1/B2 Biosafety Cabinets
The field certification of Biosafety Cabinets (BSC’s) as required in Annex F of the NSF/ANSI 49 class II bi...
Other content in this Stream
[Bulletin] Care and Use of Copper Surfaces
NuAire lab products manufactured with copper such as CO2 Incubators have specific care and use instructions to assure maximum customer satisfaction.
[Webinar] Biosafety Cabinet Introduction
Learn some of the basic about biosafety cabinets and how to go about selecting the right bsc for your application.
[Spec] NU-677 Class II, Type A2 Animal Handling Biosafety Cabinet Specification (115V)
[Spec] NU-481E Specification (230V)
[Spec] NU-481 Specification (115V)
[Spec] NU-540E Class II Microbiological Safety Cabinet (230V)
[Spec] NU-540 Class II, Type A2 Biosafety Cabinet (115V)
[Spec] NU-543E Class II Microbiological Safety Cabinet (230V)
[Spec] Class II, Type B2 Biosafety Cabinet NU-430 Specification (115V)
LabGard model NU-430 Class II, Type B2 Total Exhaust Biosafety Cabinet specification.
[Spec] Class II, Type B1 Biosafety Cabinet Model NU-427 Specification (115V)
View specifications for NSF/ANSI 49 listed LabGard Model NU-427 Class II, Type B1 Biosafety Cabinet.
[White Paper] Use of HEPA Filters in Biosafety Cabinets
Learn about the use of HEPA filters in biosafety cabinets.
[Flyer] Mini Microcentrifuges
Mini Microcentrifuges from NuAire Laboratory Equipment are available in three different models all at varying maximum speeds.
[Flyer] NuAire Research Laboratory Equipment Line Card
[Brochure] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
[Spec] NU-PR797 Specification (115v)
[Brochure] In-VitroCell CO2 Incubator
In-VitroCell CO2 Incubators from NuAire provide optimal growth conditions for your cells. Available in direct heat and water jacket with popular upgrades such as O2 control, RH Control, and more.
[Bulletin] Clarifying the Class II, Type C1 Biosafety Cabinet
The Class II, Type C1 Biosafety Cabinet (BSC) is similar to a Type B1 regarding airflow pattern. The Type C1 has a portion of the work surface to exhaust air.
[Specification] NU-5800 Series Direct Heat CO2 Incubator
Learn the technical side of the NU-5800 Series Direct Heat CO2 Incubator. Dimensions, performance, and more.
[Spec] In-VitroCell NU-5700 Series CO2 Incubator Specification
Purhcase specifciation for CO2 Incubator Series In-VitroCell NU-5700.
[Spec] Class I Containment Ventilated Enclosure NU-813 (115V, 230V)