Laboratory centrifuges are application-driven products. Laboratory professionals should be well-informed before determining which is the correct centrifuge to meet their application. In order to help identify the correct product for an application, there are seven questions to consider. These questions will help identify which product category and which model within that category is appropriate. In addition to the correct category and model, it is important to identify the accessories, rotor, buckets, and inserts that are available to expand the capabilities of a centrifuge or allow it to perform a function critical to a needed application.
[White Paper] Ultralow Freezer Buying Guide
Read this Ultralow Freezer Buying Guide to learn what questions to ask when purchasing your next ultralow t...
Other content in this Stream
[Spec] NU-481 Specification (115V)
[Spec] NU-540 Class II, Type A2 Biosafety Cabinet (115V)
[Spec] Class II, Type B2 Biosafety Cabinet NU-430 Specification (115V)
LabGard model NU-430 Class II, Type B2 Total Exhaust Biosafety Cabinet specification.
[Spec] Class II, Type B1 Biosafety Cabinet Model NU-427 Specification (115V)
View specifications for NSF/ANSI 49 listed LabGard Model NU-427 Class II, Type B1 Biosafety Cabinet.
[Brochure] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
[Brochure] In-VitroCell CO2 Incubator
In-VitroCell CO2 Incubators from NuAire provide optimal growth conditions for your cells. Available in direct heat and water jacket with popular upgrades such as O2 control, RH Control, and more.
[Specification] NU-5800 Series Direct Heat CO2 Incubator
Learn the technical side of the NU-5800 Series Direct Heat CO2 Incubator. Dimensions, performance, and more.
[Spec] Class I Containment Ventilated Enclosure NU-813 (115V, 230V)
LabGard model NU-813 is a Class I Containment Ventilated Enclosure providing personnel protection for handling low-risk aerosols and particulate matter.
[White Paper] Centrifuge Maintenance: Protecting the Investment
Centrifuges must be operated correctly and undergo regular maintenance in order to achieve maximum lifespan.
[Brochure] Hitachi CR22N High Speed Centrifuge
[Brochure] Hitachi FNX Micro Ultracentrifuge
[Brochure] Hitachi CP-WX Ultracentrifuge
Modify A NuWind Centrifuge Program
Updating or adding a NuWind Centrifuge program can be achieved by selecting the proper icon with your finger located on the touchscreen controls and clicking save.
How To Change A NuWind Centrifuge Rotor Gasket
Gaskets located on a NuWind Centrifuge rotor help to create a secure bond between the lid and the body of the rotor. Replacing a gasket is part of preventative maintenance that will assist as a safety
How To Install A NuWind Centrifuge Bucket Lid
The bucket of a NuWind centrifuge features a cam action type lock to securely create a biosafety seal between the bucket and lid.
How To Change A NuWind Centrifuge Rotor
Changing a rotor on a NuWind Centrifuge can be done in as little as 10 seconds. A twist of a dial, removal of a rotor, placement of a new rotor, an audible "Click" you are done.
How To Install A NuWind Centrifuge Rotor
Removing and installing a rotor on a NuWind centrifuge is as simple as a twist of a dial and listening for an audible "Click".
How To Grease Your Centrifuge
Preventative maintenance such as greasing your centrifuge on a regular basis ensures a long product life and better results.
Initial Startup of a NuWind Centrifuge
Just unpacked your NuWind Centrifuge? Learn the basic setup instructions to power on your new Centrifuge.
[Bulletin] NU-480 Decontamination by Vaporized Hydrogen Peroxide