Home » General Technical Bulletins » [Bulletin] NU-NTE797 Main Chamber Pressure Specification Change
Most Recent Flipbooks
[Bulletin] Care and Use of Copper Surfaces
NuAire lab products manufactured with copper such as CO2 Incubators have specific care and use instructions to assure maximum customer satisfaction.
[Bulletin] Clarifying the Class II, Type C1 Biosafety Cabinet
The Class II, Type C1 Biosafety Cabinet (BSC) is similar to a Type B1 regarding airflow pattern. The Type C1 has a portion of the work surface to exhaust air.
[Bulletin] NuAire Laboratory Equipment Electrical Safety
This technical bulletin is intended review both UL and NFPA requirements and clarifies the requirements for non-patient care laboratyr and pharmacy equipment.
[Bulletin] Volatile Contaminant Air Concentrations in Type B Biosafety Cabinets
Use of volatiles in Class II, Type B Biosafety Cabinets (BSCs).
[Bulletin] Camera Usage in a Biosafety Cabinet and Pharmacy Isolator
NuAire offers general guidance to aid in the selection of camera systems that can be mounted on or near your primary engineering controls.
[Bulletin] Impact of Class II Biosafety Cabinet Downflow Velocity on Cross Contamination
The result of lower Biosafety Cabinet downflow velocities result in a higher level of sample cross-contamination using traditional work practice guidelines.
[Bulletin] NU-S125 Performance Evaluation
[Bulletin] NU-480 Decontamination by Vaporized Hydrogen Peroxide
[Bulletin] NU-480-600E Decontamination Process Testing Results
[Bulletin] Concurrent Balance Value of Class II, Type B1/B2 Biosafety Cabinets
The field certification of Biosafety Cabinets (BSC’s) as required in Annex F of the NSF/ANSI 49 class II biosafety cabinetry standard is based on the principle of replication.
[Bulletin] Movement and Re-certification of a Biosafety Cabinet
NU-NTE800 Performance Evaluation Compliance to USP 797 / USP 800
[Bulletin] Use of Anesthetic Gas Isoflurane within a Biosafety Cabinet
Surgical procedures using Anesthetic Gas in the vivarium that require a sterile or contained environment should be performed in a biosafety cabinet (BSC).
NU-NR800 Performance Evaluation Compliance to USP 797 / USP 800
[Bulletin] When to Replace My Biosafety Cabinet
Biosafety Cabinets have a 15 year useable life. Read this bulletin to answer the question when to replace your biosafety cabinet.
NU-PR797 Performance Evaluation Compliance to USP 797
[Bulletin] Chemo Pad Containment Capability of a Biosafety Cabinet
Performance capabilities of a NuAire biosafety cabinet (BSC) using a chemotherapy pad on the worksurface.
Case and Use of Polypropylene
NU-603 Reverse Flow Cabinet Performance Evaluation
[Bulletin] Natural Gas use in a Biosafety Cabinet