Biosafety Cabinets (BSC’s) are designed to offer protection to the end user, the materials you are working on, as well as the environment. They are found in a variety of BSL (Biosafety Level) laboratories. The right BSC can offer greater protection than others but in the end, if the cabinet is not used with proper techniques than protection is lost and contamination becomes a threat. Here are a few simple tips on how to work safely in your biosafety cabinet.
[Tip] Never Cover the Airflow Grill on a Biosafety Cabinet
The front airflow grill on a biological safety cabinet performs the important job of separating the clean a...
Shop Biosafety CabinetsView Now
Other content in this Stream
[Warranty] Class II, Type A2 Biosafety Cabinet (International)
[Warranty] Class II, Type B2 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type B1 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type A2 Biosafety Cabinet (US & CA)
Multiple Users in a Class II Biosafety Cabinet
This is an age old question that comes every now and then. First in reference from the BMBL (Biosafety in Microbiological and Biomedical Laboratories), a clear 12-inches (305 mm) should be present...
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Flyer] LabGard NU-565 Class II, Type B2 Biosafety Fume Hood
Learn more about the LabGard Class II, Type B2 Biosafety Fume Hood model NU-565.
[Flyer] LabGard NU-560 Class II, Type B2 Biosafety Cabinet
Learn more about the Total Exhaust LabGard Class II, Type B2 Biosafety Cabinet (BSC) model NU-560.
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Blog] Transitioning from Biosafety Cabinet models NU-425 and NU-475 to model NU-540
NuAire biosafety cabinet model NU-540 has replace models NU-425 and NU-475. Learn what the differecnes are when transitioning to model NU-540.
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Infographic] Telescoping Base Stand Assembly
Learn how to properly assemble a telescoping base stand from NuAire to fit anyone of our Biosafety Cabinets (BSC), Laminar Airflow Workstations (LAFW), and more.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet
[Webinar] Biosafety Cabinet Introduction