In today’s world, residential, commercial and institutions (homes, businesses, stores, and schools) commonly install camera surveillance systems for the protection of occupants and visitors. Recently with the news of incidents within pharmacies where the Compounded Sterile Products (CSP’s) were not prepared properly (compromised sterility, incorrect ingredients or concentration) there has been an increased awareness for the use of monitoring or surveillance systems to be located on or near the Primary Engineering Controls (PEC’s) for the purpose of monitoring the very critical work and/or maintaining video evidence on past procedures and training. To accomplish this, NuAire is now offering general guidance to aid in the selection of camera systems that can be mounted on or near the PEC for both new PEC purchases and for field retrofit. In addition to the general guidance, NuAire offers one camera choice along with accessories (i.e. cabling, monitors, DVR’s) for a turnkey solution. Nuaire, as in the past, offers special installation options for camera systems that are provided by the customer. For further information, please contact your NuAire representative, distributor or factory sales team member for assistance.
[Bulletin] Volatile Contaminant Air Concentrations in Type B Biosafety Cabinets
Use of volatiles in Class II, Type B Biosafety Cabinets (BSCs).
Shop Biosafety CabinetsView Now
Other content in this Stream
[Flyer] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
[Warranty] Class II, Type A2 Biosafety Cabinet (International)
[Warranty] Class II, Type B2 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type B1 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type A2 Biosafety Cabinet (US & CA)
Multiple Users in a Class II Biosafety Cabinet
This is an age old question that comes every now and then. First in reference from the BMBL (Biosafety in Microbiological and Biomedical Laboratories), a clear 12-inches (305 mm) should be present...
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Flyer] LabGard NU-565 Class II, Type B2 Biosafety Fume Hood
Learn more about the LabGard Class II, Type B2 Biosafety Fume Hood model NU-565.
[Flyer] LabGard NU-560 Class II, Type B2 Biosafety Cabinet
Learn more about the Total Exhaust LabGard Class II, Type B2 Biosafety Cabinet (BSC) model NU-560.
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Blog] Transitioning from Biosafety Cabinet models NU-425 and NU-475 to model NU-540
NuAire biosafety cabinet model NU-540 has replace models NU-425 and NU-475. Learn what the differecnes are when transitioning to model NU-540.
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Infographic] Telescoping Base Stand Assembly
Learn how to properly assemble a telescoping base stand from NuAire to fit anyone of our Biosafety Cabinets (BSC), Laminar Airflow Workstations (LAFW), and more.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet