NSF/ANSI 49 and BMBL (Biosafety in Microbiological and Biomedical Laboratories) reference the term “Relocated” as the measure of requiring re-certification. Vertical up and down BSC movement on a motorized base stand is not relocation as intended by either of these documents. The height adjustment up and down movement does not cause the vibration necessary to harm or breach the integrity of the HEPA filters. The height adjustment up and down movement is a very slow and smooth movement typical driven by a hydraulic system. If the height adjustment is used in conjunction with an exhausted BSC (i.e. type B1/B2 or canopy connected type A2), the site assessment certification test should include airflow verification throughout the range of motion. The duct connection itself should be made with flexible PVC duct material.
[Bulletin] Concurrent Balance Value of Class II, Type B1/B2 Biosafety Cabinets
The field certification of Biosafety Cabinets (BSC’s) as required in Annex F of the NSF/ANSI 49 class II bi...
Shop Biosafety CabinetsView Now
Other content in this Stream
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Instructions] Telescoping Base Stand Assembly
Learn how to assemble your NuAire telescoping base stand for your biosafety cabinet, containment ventilated enclosure, and more.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet
[Webinar] Biosafety Cabinet Introduction
Learn some of the basic about biosafety cabinets and how to go about selecting the right bsc for your application.
[Flyer] LabGard NU-545E Class II Biosafety Cabinet (230V)
Product flyer for LabGard AIR Limited 230-volt Class II Biosafety Cabinet (BSC) model NU-545E.
[Flyer] LabGard NU-545 Class II, Type A2 Biosafety Cabinet (115V)
Product flyer for LabGard AIR Limited Class II, Type A2 Biosafety Cabinet (BSC) model NU-545.
[Video] Work Safely in your Biosafety Cabinet
Learn the basics of how NuAire Biological Safety Cabinets are designed to offer optimal personnel, environmental and product protection.
[Infographic] Biosafety Cabinet Decommissioning Process
Learn the Biosafety Cabinet Decommissioning Process with this infographic.
[Bulletin] Biosafety Cabinet Seismic Anchorage Calculations & Installation
Seismic anchorage calculations and installation instructions for fixed NuAire Biosafety cabinets.
[Ad] The Safe in Biosafety Cabinet
LabGard BSC's from NuAire Lab Equipment put the Safe in Biosafety Cabinet with a long lasting monolithic design.
Ergonomically Designed Biosafety Cabinet
Reduce workplace errors and injuries with an ergonomically designed LabGard Biosafety Cabinet from NuAire Laboratory Equipment.
[Report] Containment Ventilated Enclosure (CVE) Containment Test
View performance results on Containment Ventilated Enclosure (CVE) model NU-813 while compounding Estradiol capsules.
[Spec] NU-677E Class II Animal Handling Biosafety Cabinet Specification (230V)
[Spec] NU-677 Class II, Type A2 Animal Handling Biosafety Cabinet Specification (115V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)