Laboratory research and pharmacy compounding inside a biological safety cabinet (BSC) are tedious work. Proper laboratory ergonomics is a must for laboratory personnel. A properly design BSC can help minimize stress resulting in better research results. If you have an existing BSC adding simple accessories such as a footrest or elbow rest can make a huge difference. Read these ten ergonomic tips for working in your biosafety cabinet to help reduce workplace injuries.
LabGard® ES AIR NU-543E Class II Biological Safety Cabinet
Model NU-543E is a Class II Biosafety Cabinet featuring TUV and NSF listings at a 254 mm (10-inch) access o...
Shop Biosafety CabinetsView Now
Other content in this Stream
[Warranty] Class II, Type A2 Biosafety Cabinet (International)
[Warranty] Class II, Type B2 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type B1 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type A2 Biosafety Cabinet (US & CA)
Multiple Users in a Class II Biosafety Cabinet
This is an age old question that comes every now and then. First in reference from the BMBL (Biosafety in Microbiological and Biomedical Laboratories), a clear 12-inches (305 mm) should be present...
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Flyer] LabGard NU-565 Class II, Type B2 Biosafety Fume Hood
Learn more about the LabGard Class II, Type B2 Biosafety Fume Hood model NU-565.
[Flyer] LabGard NU-560 Class II, Type B2 Biosafety Cabinet
Learn more about the Total Exhaust LabGard Class II, Type B2 Biosafety Cabinet (BSC) model NU-560.
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Blog] Transitioning from Biosafety Cabinet models NU-425 and NU-475 to model NU-540
NuAire biosafety cabinet model NU-540 has replace models NU-425 and NU-475. Learn what the differecnes are when transitioning to model NU-540.
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Infographic] Telescoping Base Stand Assembly
Learn how to properly assemble a telescoping base stand from NuAire to fit anyone of our Biosafety Cabinets (BSC), Laminar Airflow Workstations (LAFW), and more.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet
[Webinar] Biosafety Cabinet Introduction