Laboratory research and pharmacy compounding inside a biological safety cabinet (BSC) are tedious work. Proper laboratory ergonomics is a must for laboratory personnel. A properly design BSC can help minimize stress resulting in better research results. If you have an existing BSC adding simple accessories such as a footrest or elbow rest can make a huge difference. Read these ten ergonomic tips for working in your biosafety cabinet to help reduce workplace injuries.
Shop Biosafety CabinetsView Now
Other content in this Stream
[Flyer] LabGard NU-565 Class II, Type B2 Biosafety Fume Hood
Learn more about the LabGard Class II, Type B2 Biosafety Fume Hood model NU-565.
[Flyer] LabGard NU-560 Class II, Type B2 Biosafety Cabinet
Learn more about the Total Exhaust LabGard Class II, Type B2 Biosafety Cabinet (BSC) model NU-560.
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Blog] Transitioning from Biosafety Cabinet models NU-425 and NU-475 to model NU-540
NuAire biosafety cabinet model NU-540 has replace models NU-425 and NU-475. Learn what the differecnes are when transitioning to model NU-540.
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet
[Webinar] Biosafety Cabinet Introduction
Learn some of the basic about biosafety cabinets and how to go about selecting the right bsc for your application.
[Flyer] LabGard NU-545E Class II Biosafety Cabinet (230V)
Product flyer for LabGard AIR Limited 230-volt Class II Biosafety Cabinet (BSC) model NU-545E.
[Flyer] LabGard NU-545 Class II, Type A2 Biosafety Cabinet (115V)
Product flyer for LabGard AIR Limited Class II, Type A2 Biosafety Cabinet (BSC) model NU-545.
[Video] Work Safely in your Biosafety Cabinet
Learn the basics of how NuAire Biological Safety Cabinets are designed to offer optimal personnel, environmental and product protection.
[Infographic] Biosafety Cabinet Decommissioning Process
Learn the Biosafety Cabinet Decommissioning Process with this infographic.
[Bulletin] Biosafety Cabinet Seismic Anchorage Calculations & Installation
Seismic anchorage calculations and installation instructions for fixed NuAire Biosafety cabinets.
[Ad] The Safe in Biosafety Cabinet