Over the last several years, Biosafety Cabinet (BSC) manufacturers have lowered downflow velocity while maintaining inflow velocity to increase energy efficiency, reduce exhaust system requirements and lower BSC Noise and vibration levels. However, the lowering of downflow velocity may come with a consequence of greater lateral airflow movement within the work zone creating more significant potential for cross-contamination. The lowering of the downflow velocity does not compromise personnel and product containment performance. Why? Primarily, by maintaining inflow velocity above a certain level as required by the standard, typically 100 fpm (.51 mps). However, the standard does not require a specific downflow velocity and can be adjusted as needed based on the design parameters of the work zone supply diffuser and grill patterns (airflow distribution) to assure compliance to the personnel and product protection requirements of NSF/ANSI 49. NSF/ANSI 49 test procedures do test for cross-contamination, but only at the work zone sidewalls, which doesn’t necessarily translate into the lateral movement of airflow in the center of the work zone. The result of these lower downflow velocities may be a higher level of sample cross-contamination using traditional work practice guidelines.
Home » Biosafety Cabinets » [Bulletin] Impact of Class II Biosafety Cabinet Downflow Velocity on Cross Contamination
Shop Biosafety CabinetsView Now
Other content in this Stream
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Flyer] LabGard NU-565 Class II, Type B2 Biosafety Fume Hood
Learn more about the LabGard Class II, Type B2 Biosafety Fume Hood model NU-565.
[Flyer] LabGard NU-560 Class II, Type B2 Biosafety Cabinet
Learn more about the Total Exhaust LabGard Class II, Type B2 Biosafety Cabinet (BSC) model NU-560.
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Blog] Transitioning from Biosafety Cabinet models NU-425 and NU-475 to model NU-540
NuAire biosafety cabinet model NU-540 has replace models NU-425 and NU-475. Learn what the differecnes are when transitioning to model NU-540.
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet
[Webinar] Biosafety Cabinet Introduction
Learn some of the basic about biosafety cabinets and how to go about selecting the right bsc for your application.
[Flyer] LabGard NU-545E Class II Biosafety Cabinet (230V)
Product flyer for LabGard AIR Limited 230-volt Class II Biosafety Cabinet (BSC) model NU-545E.
[Flyer] LabGard NU-545 Class II, Type A2 Biosafety Cabinet (115V)
Product flyer for LabGard AIR Limited Class II, Type A2 Biosafety Cabinet (BSC) model NU-545.
[Video] Work Safely in your Biosafety Cabinet
Learn the basics of how NuAire Biological Safety Cabinets are designed to offer optimal personnel, environmental and product protection.
[Infographic] Biosafety Cabinet Decommissioning Process
Learn the Biosafety Cabinet Decommissioning Process with this infographic.
[Bulletin] Biosafety Cabinet Seismic Anchorage Calculations & Installation
Seismic anchorage calculations and installation instructions for fixed NuAire Biosafety cabinets.
[Ad] The Safe in Biosafety Cabinet