It happens at some point to even the most seasoned laboratory technician that a spill occurs within the Biosafety Cabinet (BSC). Taking precautionary measures before and during your work with hazardous materials could help keep you and others safe. Remember, if a spill occurs, don’t panic. Here are some simple steps to keep you and your laboratory safe. Please check with your Environmental Health and Safety (EHS) office or Biosafety Officer to ensure you have proper steps in place in case of a spill based of standard Biosafety in Microbiological and Biomedical Laboratories (BMBL).
Shop Biosafety CabinetsView Now
Other content in this Stream
[Warranty] Class II, Type A2 Biosafety Cabinet (International)
[Warranty] Class II, Type B2 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type B1 Biosafety Cabinet (US & CA)
[Warranty] Class II, Type A2 Biosafety Cabinet (US & CA)
Multiple Users in a Class II Biosafety Cabinet
This is an age old question that comes every now and then. First in reference from the BMBL (Biosafety in Microbiological and Biomedical Laboratories), a clear 12-inches (305 mm) should be present...
[Spec] NU-560E Class II, Type B2 Biosafety Cabinet (230V)
[Spec] NU-560 Class II, Type B2 Biosafety Cabinet (115V)
[Spec] LabGard NU-565 Series 1 Class II, Type B2 Biosafety Cabinet
[Flyer] LabGard NU-565 Class II, Type B2 Biosafety Fume Hood
Learn more about the LabGard Class II, Type B2 Biosafety Fume Hood model NU-565.
[Flyer] LabGard NU-560 Class II, Type B2 Biosafety Cabinet
Learn more about the Total Exhaust LabGard Class II, Type B2 Biosafety Cabinet (BSC) model NU-560.
[Infographic] 10 Tips for Working Safely in the Pharmacy with Your Biosafety Cabinet
Ten tips to help you use the proper aseptic technique when working within a Class II Biosafety Cabinet (BSC) to minimize air turbulence and prevent the spread of aerosols.
[Blog] Transitioning from Biosafety Cabinet models NU-425 and NU-475 to model NU-540
NuAire biosafety cabinet model NU-540 has replace models NU-425 and NU-475. Learn what the differecnes are when transitioning to model NU-540.
[Infographic] Class II Biosafety Cabinet Assembly
Upon arrival of your Class II Biosafety Cabinet follow these steps for the basic assembly of your cabinet.
[Infographic] Telescoping Base Stand Assembly
Learn how to properly assemble a telescoping base stand from NuAire to fit anyone of our Biosafety Cabinets (BSC), Laminar Airflow Workstations (LAFW), and more.
[Infographic] Class II, Type A2 Biosafety Cabinet Placement
Learn the ideal location to install your Class II, Type A2 Biosafety Cabinet to maximize safety and performance.
[Drawing] NU-560, NU-565 Class II, Type B2 Biosafety Cabinet Airflow Schematic
[Guide] CETA Application Guide for the Exhaust System Requirements of Class II, Type B Biosafety Cabinets
Learn the about Class II, Type B1 and B2 Biosafety Cabinet exhaust requirements with this applicaiton guide from CETA International.
[Drawing] NU-560-600 Class II, Type B2 Biosafety Cabinet
[Drawing] NU-560-400 Class II, Type B2 Biosafety Cabinet
[Webinar] Biosafety Cabinet Introduction