Chapter 2: Understanding the Biosafety Cabinet
Biological Safety Cabinets were designed to remove or minimize exposures to hazardous biological materials ...
Other content in this Stream
Clarifying the Type C1 Biosafety Cabinet
Biosafety Cabinet Operating Label
Delivery and Setup for your Pharmacy Isolator
Watch what items you should check off your list once your pharmacy isolator arrives at your loading dock.
[Infographic] CACI Sleeve and Glove Changing Procedure
[Infographic] CAI Sleeve and Glove Changing Procedure
[Infographic] 5 Biosafety Cabinet Reminders
Hazardous Drug Biosafety Cabinet Operating Label
[Infographic] 10 Tips for Working in Your Biosafety Cabinet
10 Steps to Cleaning a Spill in a Biosafety Cabinet
Facility and Engineering Controls Using USP 800 Guidelines
Changing Gloves in a CACI
Changing gloves on your Compounding Aseptic Containment Isolator must be done on a routine basis. Follow these simple steps to safely change your CACI gloves.
Airflow in a Negative Pressure Compounding Aseptic Containment Isolator
Airflow in a negative pressure compounding aseptic containment isolator (CACI) is designed to keep you safe.
Testing Your Adjustable Base Stand
Testing the adjustable base stand of your new pharmacy isolator is a great way to ensure your product is workly correctly.
First Air Sterility
Maintain sterility in your PharmaGard isolator by avoiding disrupting air over items located on the work surface.
LabGard NU-500 Series Control Systems
Work in a safer environment with NuAire electronics monitoring and controlling your work environment.
LabGard NU-500 Series Ergonomics
Minimize workplace injuries working the comfort of a LabGard Biosafety Cabinet.
LabGard NU-500 Series Construction
Feel at ease working in a longer lasting safer environment.
Introducing the LabGard 500 Series Biosafety Cabinet
The LabGard NU-500 Series Class II Biosafety Cabinets from NuAire provide you the very best in design, ergonomics, and technology.
Camera Useage Information for Biosafety Cabinets and Pharmacy Isolators
Impact of Class II Biosafety Cabinet Downflow Velocity on Cross Contamination