Learn steps to take before you begin work in your Biological Safety Cabinet. A proper biosafety or operations manuals will specify practices and procedures designed to minimize and eliminate risks.
Chapter 4: Using Proper Procedures and Techniques
The use of proper procedures and equipment cannot be overemphasized in providing primary personnel and envi...
Shop Biosafety CabinetsView Now
Other content in this Stream
[Spec] NU-481 Specification (115V)
[Brochure] LabGard AIR NU-543 Class II, Type A2 Biosafety Cabinet
LabGard AIR Model NU-543 Class II, Type A2 Biosafety Cabinet provides long lasting personnel, product, and environmental protection through advanced construction techniques and electronic controls.
Clarifying the Type C1 Biosafety Cabinet
NU-813 Specification (115V, 230V)
[White Paper] Can a Class I and Class II BSC be in the same Sterile Hazardous Drug (HD) Room?
USP 800 does not directly address the pre-sterilization process of a sterile preparation creating the question can a class I and a Class II biosafety cabinet be used in the same sterile HD room?
NuAire Laboratory Equipment Electrical Safety
LabGard ES NU-813 Class I Biosafety Cabinet for Pharmacy
[Infographic] Biosafety Cabinet Ergonomic Tips
Biosafety Cabinet Operating Label
[White Paper] Benefits and Risks of Ultraviolet (UV) Light
Biosafety Cabinet Selection Guide
[Brochure] LabGard NU-677 Animal Handling Class II, Type A2 Biosafety Cabinet
[Infographic] 5 Biosafety Cabinet Reminders
Hazardous Drug Biosafety Cabinet Operating Label
[Infographic] 10 Tips for Working in Your Biosafety Cabinet
Variable Airflow (VFC) Exhaust Canopy
AireValv Butterfly Valves for Exhausting
10 Steps to Cleaning a Spill in a Biosafety Cabinet
LabGard NU-500 Series Control Systems
Work in a safer environment with NuAire electronics monitoring and controlling your work environment.
LabGard NU-500 Series Ergonomics